Spare Parts and Accessories
Feedline accessory kit (1/pkg)
VacuCap 60 Devices
0.1 µm, 60 mm, gamma-irradiated (10/pkg)
0.2 µm, 60 mm, gamma-irradiated (10/pkg)
0.45 µm, 60 mm, gamma-irradiated (10/pkg)
0.2 µm, 60 mm (supplied with individually attached tubing for each filter device), gamma-irradiated (10/pkg)
VacuCap 60 PF Device
0.8/0.2 µm, 60 mm, gamma-irradiated (10/pkg)
VacuCap 90 Devices
0.1 µm, 90 mm, gamma-irradiated (10/pkg)
0.2 µm, 90 mm, gamma-irradiated (10/pkg)
0.45 µm, 90 mm, gamma-irradiated (10/pkg)
VacuCap 90 PF Device
0.8/0.2 µm, 90 mm, gamma-irradiated (10/pkg)
View our video on vacuum filtration made easy
Innovative Bottle-Top Filters for Fast Vacuum Filtration of 100 mL to 5 L of Aqueous Solutions
Vaccum Filtration Made Easy
Explore how VacuCap filter unit can:
- Replace current vacuum filter systems
- Simplify workflow
- Maximize filtration
- Filter faster. Supor® membrane provides high flow rates.
- No need to constantly refill upper fluid reservoir.
- Draws directly from the mixing reservoir.
- Eliminates the possibility of contamination from transfer steps. Filters directly into the desired container.
- Match filter to sample volume by choosing from two sizes.
- Available built-in prefilter increases throughput of particulate-laden solutions such as serum-containing media.
- Environmentally-friendly with minimal plastic waste.
- Patented small design accepts a variety of collection vessels and reduces storage space and waste.
- TA version available with tubing attached to each device for maximum convenience.
- Gamma irradiated, eliminates potential contamination by EtO residuals.
View our video on vacuum filtration made easy
- Excellent device for cell culture media preparation
- PF version is useful for sterilization of hard-to-filter solutions
- Reduce mycopalsma with the 0.1 µm pore size
Materials of Construction
- Filter Media: Supor [hydrophilic polyethersulfone (PES)] membrane
- Housing: Modified acrylic
- Membrane Support Material: Polyester
- Sinker Material: Glass-filled polyurethane elastomer
- Inlet Tubing: Polyvinyl chloride (PVC) medical-grade tubing
- Gasket Seal Material: Polyethylene
- 0.1, 0.2, 0.45, and 0.8/0.2 μm
Effective Filtration Area
- VacuCap 60 Devices: 30 cm2
- VacuCap 90 Devices: 60 cm2
- (RPMI + 10% newborn calf serum)
VacuCap 60 Devices, 0.2 μm: 1 L
VacuCap 90 Devices, 0.2 μm: 5 L
- (RPMI + 10% calf serum)
VacuCap 60 PF Devices: 500 mL
VacuCap 90 PF Devices: 1 L
- VacuCap 60 Devices: Can be used on receptacles with openings ranging from 2 - 5 cm (0.8 - 1.9 in.)
- VacuCap 90 Devices: Can be used on receptacles with openings ranging from 2 - 6.5 cm (0.8 - 2.5 in.)
Typical Hold-Up Volume
- Vacucap 60 Devices: 1.2 mL
- VacuCap 90 Devices: 3.2 mL
Maximum Operating Temperature
- 55 °C (131 °F)
Typical Water Flow Rate
- mL/min at 25.4 cm Hg (10 in. Hg)
VacuCap 60 Devices VacuCap 90 Devices 0.1 µm: 50 0.1 µm: 100 0.2 µm: 200 0.2 µm: 400 0.45 µm: 280 0.45 µm: 560 PF: 200 PF: 400
- 38.1 cm Hg (15 in. Hg)
- < 0.25 EU/mL using Limulus Amoebocyte Lysate (LAL) test
- Passes United States Pharmacopeia (USP) Biological Reactivity Test, In Vivo <88>
- Sterilized by gamma irradiation and individually packaged
VacuCap® 90 Device Filtration Efficiency(RPMI 1640 + 10% Newborn Calf Serum)
VacuCap 90 Device Filtration Capacity(100% Newborn Calf Serum)
VacuCap 90 (0.2 μm) filter device performance as compared to that of competitive vacuum filter units with 0.2 μm Cellulose Acetate membrane. Filtration conducted at 21 °C (70 °F) and 38 cm Hg (15 in. Hg) vacuum. Volumes reflect a filtration time of five minutes.
VacuCap 90 PF Device Filtration Efficiency(RPMI + 10% Calf Serum)
VacuCap 90 PF Device Filtration Capacity(100% Calf Serum)
Filtration conducted at 21 °C (70 °F), 50 cm Hg (20 in. Hg) vacuum, five minute filtration time. Actual results may vary depending upon type and concentration of serum, liquid temperature, and applied vacuum.
- Connect the feed tubing to the port marked “INLET” on the VacuCap device. Place the opposite end of the tubing in the unfiltered fluid to be drawn.
- Connect the vacuum tubing to the port marked “VACUUM” on the VacuCap device. Refer to product insert for safety precautions.
- While holding the VacuCap device securely onto the filtrate container, start the vacuum. The VacuCap device will seal securely to the container top and fluid will be drawn.
- When filtration is complete, switch off the vacuum pump allowing the vacuum inside the receiving container to dissipate. Refer to the product insert for complete instructions.
Vacuum Filtration Device
Sterile Filtration and Clarification