The Kleenpak sterile disconnector provides an easy to use, secure method for the sterile disconnection of flexible tubing assemblies, even in an uncontrolled environment. It can also be used in processes involving single-use systems as well as hybrid systems. The disconnector maintains sterility of the fluid path before and after disconnection.
Product images are a representation of product family for illustration purposes only. Actual products may vary.
The disconnector consists of a joined male and female unit, to fit tubing with 13 mm (½ in.) ID. Kleenpak sterile disconnectors can either be autoclaved up to 130 °C or gamma irradiated up to 50 kGy.
The Kleenpak sterile disconnector brings a major improvement to single-use technology in aseptic processing. It can be used at all stages of the process, but will be particularly valuable in upstream and formulation & filling, where both ease of use and sterility assurance are critical. Pall can engineer and manufacture state-of-the-art Allegro™ single-use systems with Kleenpak sterile disconnectors for all these critical applications.
Warning: This product is not sold sterile. For use in making sterile disconnections, each disconnector must be fitted to a closed single-use assembly that has been subjected to a validated sterilizing process.
|Component||Material of Construction|
|Male side plunger||Polysulfone|
|Female side connector||Polysulfone|
|Anti-actuation collar||Thermoplastic elastomer|
|Anti-disconnection collar||Thermoplastic elastomer|
|Female end cover||Polysulfone|
|Water at 60-65 °C after autoclaving at 130 °C and Gamma irradiation of 51.2 kGy||Typically < 1 mg|
|Gamma irradiation||Maximum 50 kGy|
|Autoclave||75 minutes at 130 °C|
For additional information about sterilization methods, please contact Pall.
|Maximum operating pressure and temperature||3 bar at 40 °C|
|6 mm (¼ in.)||9 mm (3⁄8 in.)||
13 mm (1⁄2 in.)
|Total length (A)||156 mm (6.1 in.)||156 mm (6.1 in.)||121 mm (4.8 in.)|
|Maximum diameter (B)||61 mm (2.4 in.)||61 mm (2.4 in.)||61 mm (2.4 in.)|
|Internal diameter (C)||6.35 mm (0.25 in.)||9.5 mm (0.38 in.)||13 mm (0.5 in.)|
|Maximum diameter male side (D)||61 mm (2.4 in.)||61 mm (2.4 in.)||61 mm (2.4 in.)|
|Maximum length male side once disconnected (E)||102 mm (4.0 in.)||102 mm (4.0 in.)||85 mm (3.3 in.)|
|Maximum length female sidewith female end cover once disconnected (F)||52 mm (2.0 in.)||52 mm (2.0 in.)||36 mm (1.4 in.)|
|Maximum diameter of female side with female end cover (G)||36 mm (1.4 in.)||36 mm (1.4 in.)||36 mm (1.4 in.)|
Male Side - Disconnected
Female Side with Cap - Disconnected
1For illustrative purposes, drawings show 13 mm (1⁄2 in.) hose barb connections
Step 1: Pull (remove) male side anti-actuation collar
Step 2: Push male section into female sections 'CLICK'
Step 3: Pull (remove) anti-disconnection collar
Step 4: Twist to separate the two sections